I02 is one of five macromolecular crystallography (MX) beamlines at Diamond. I02 is a tuneable beamline with a working wavelength range of 0.68 - 2.48 Å. The standard working energy is 12.658 keV which provides a wavelength of 0.979 Å and a focussed beam size of 80 x 20 um (FWHM). As with all the MX beamlines, I02 uses SPINE standard pins and Unipucks.
MX bag traing registration open until 13th June. Register here
Remote access available on all MX beamlines at all times since January 2013
428 structures have been published from I02: 21 so far structures in 2013.
I02 will take first users after the shut down on Friday 21st April.
|Detector type||Pilatus 6M|
|Standard working wavelength (Å / keV)||0.979 / 12.658|
|Maximum resolution (Å) at this wavelength||1.09|
|Available wavelengths (Å)||0.5 - 2.5|
|Available wavelengths (keV)||5.0 - 18.0 (up to 25 upon request)|
|Maximum attainable resolution (Å)||0.8|
|Flux Details||Measured through 200 micron aperture @ 12658 eV and 300 mA ring current|
|Absorption edge scans||Optimised SAD and MAD experiments|
|Focussed Beam Size (µm)||80 x 15 (FWHM)|
|Slit Size (µm)||200x200 to 25x25|
|Sample changer||Rigaku ACTOR with unipucks - exchange time < 40 sec|
|Pins and Pucks||SPINE standard pins and Unipucks|
Humidity Controller for room temperature data collection (HC1)
|Available on request - now integrated into endstation setup|
Please discuss your requirements with a member of the beamline team before your experiment.
© Diamond Light Source 2012 Ltd. | Legal Notices